Ultraviolet Germicidal Lamp Holder
Ultraviolet Germicidal Lamp Holder has two pcs of 30 watts UV lamps that emit a high intensity ultraviolet light that hits every surface in the room and kills any micro organisms or pathogen including MRSA,
hand foot mouth disease,colds and flu,pneumonia,molds,e.coli,salmonella and similar type bacteria.
The light penetrates and destroys the actual DNA of micro organisms.
There is no known micro organism that is immune to UV-C energy.
Ultraviolet Germicidal Lamp Holder disinfects approximately up to 20 square meters room area.
There are numerous safety features bulit with a 0-120mins timer.
Ultraviolet Germicidal Lamp Holder has numerous applications not just for hospitals but can be used in schools,play areas,hotels,clinics,homes,cinemas,
buildings,food manufacturing companies and offices.
it quickly and effectively provide disinfection of all surfaces and air within minutes.
Ultraviolet Germicidal Lamp Holder is made of stainless steel,with its double-tube structure,it could be used independently,the technical parameters are:
Tube power: ≥ 30W×2,free ozone quartz tube.
2. Service life of tube: ≥6000hours, with smart sensing device.
3. After the lamp sterilizer is turned on, if there is still person in the room. the lamp will not be on.
4.During the sterilization, if a person or an animal breaks into the room, the lamp will be alarmed and off, automatically open left.
5. Ultraviolet Germicidal Lamp Holder is movable, foldable and equipped with timer equipment.
6. Time delay switch: after being turned on, the lamp will be on only after the person leaves.
7. Irradiance under the wavelength of 254nm (with a distance of 1m)≥214uw/cm2.
8.Ultraviolet Germicidal Lamp Holder is equipped with a timer equipment with the time limitation of 0-1440 min.
USAGE INSTRUCTION OF Ultraviolet Germicidal Lamp Holder
1.Install the base with location hole and reed.
2.Open the protection door.
3.Lift the lamp arm and adjust to 90,135,190 degrees.
4.Insert the plug and press ‘ON’ button ,adjust 0-120mins timer.
5.Leave the room right now.
6.After finishing sterilization,press ‘OFF’ button.
7.Adjust timer to ‘OFF’.
8.Hold down the ‘adjustable button 3’ to adjust lamp arm to the box .
9.Close the protection door.
10.Pull out the plug.
SAFETY CONSIDERATIONS OF Ultraviolet Germicidal Lamp Holder
Ø Don’t look directly at the ultraviolet light source,should be used in unmanned circumstances.
Ø Ensure the surface clean of UV lamp,or will affect the performance.
Ø Keep in the dry and clean room.Should Extend the exposure time if the room temperature is below 20 degrees Celsius or above 40 degrees Celsius.
Ø When sterilizing the surface of object,should illuminate directly within 1m between the ultraviolet light and surface.
Ø When sterilizing pet or animal room,please make sure pets don’t touch the ultraviolet sterilizer.Please protect the eyes and skins of pets with shading cloth.
Ø Open the window and ventilation for 10 minutes after sterilization.